Using the IFILL, you can provide the patient with the freedom to use as many tanks as they want while reducing or eliminating the need and cost of more home deliveries and more oxygen cylinders. Patients prescribed under 3LPM no longer need to be tethered to an oxygen concentrator and cannula inside their home. Compliance and Ambulation can be encouraged.
Our systems include 2 cylinders and a conserving device or regulator, so that a patient can fill one tank, while using the other.
Choose the cylinder package that fits the patients need, based on the hours of portable 02 they need per day, liter flow, and conserver tolerance.
Contact your VMS Sales Rep to get an inservice and learn more about how this technology can free your home oxygen patients while reducing the costs to your facility.
Oxygen Supply Times Chart
The chart below reflects the number of hours an oxygen-filled cylinder will last when used with a PulseDose® Conserving Device or a Continuous Flow Regulator. Note: All ambulatory ranges are calculated assuming a breath rate of 20 breaths per minute in PulseDose (PD) mode.
M-6 Cylinder 164 Gaseous Liters
“D” Cylinder 415 Gaseous Liters
“E” Cylinder 682 Gaseous Liters
The DeVilbiss iFill Personal Oxygen Station is the only stand-alone oxygen filling station on the market. With the unique design, a patient can use and store his/her iFill virtually anywhere in his/her home, minimizing disruption to the patient’s life. Additionally, the iFill can be used in conjunction with any concentrator.
Suggested Process for transfilling at home:
1. Each night, fill a tank while youre sleeping and breathing on your stationary concentrator.
2. When you wake up in the morning, swap and fill up your second cylinder while you do your morning routine.
3. You now have 2 full cylinders to start your day.
3. During the day, put any empty cylinder on the IFILL, and use your other cylinder.
4. Swap empty for full as often as you wish!
Variety of iFill Cylinders
iFill provides the flexibility to fill most cylinder sizes including M4, M6, ML6, C, D and E, using the PulseDose Oxygen Conserving Device interface or Regulator interface. iFill cylinders are also available with a standard 870 Post Valve in sizes M4, M6, ML6, C and D for use with any conserving device or regulator. (Note: iFill must be used with an iFill Cylinder.)
iFill lets users fill their cylinders in just a little over an hour – not the two hours it can take with other in-home oxygen filling units. Shorter fill times save energy costs. Partially filled cylinders can be easily topped off. And the iFill will automatically shut off when filling is complete.
Product Weight 66 lbs.
Product Dimensions 28.5” H x 12.25” W x 22.5”
Oxygen Purity 93% +/- 3%
Sound Level 50 dBA
Filling Time 75 Minutes (M6 Cylinder)
Electrical Requirements 115V, 60 Hz, 4.1 Amps
Power Consumption 400 Watts Avg.
Cylinder Top Off Yes
Warranty 3 years or 8000 hours –
whichever comes first